Confirm manual entries with ENTER. Exit the parameter level at any time by repeatedly pressing the. “Parameter” key. BioPhotometer – Short instructions. Get Eppendorf BioPhotometer Manual. Get all Eppendorf manuals!. Get Eppendorf BioPhotometer plus Instruction Manual. Get all Eppendorf manuals!.
|Published (Last):||9 September 2007|
|PDF File Size:||1.93 Mb|
|ePub File Size:||2.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
This evaluation mode is preprogrammed ex factory. BioPhotometer plus — Operating manual 2 Product description Biophtoometer BioPhotometer plus converts the measured absorbance values into concentration results. In this function list and any sub-lists e.
In the case of selection parameters, the programmable alternatives are method-dependent see overview in previous section. The flag is only for information. Switch on the device and, if necessary, the printer also. Aggressive chemicals may damage both the device and its accessories. The ideal Eppendorf product fulfilling the PhysioCare concept provides a comprehensive approach for the user. The molar mass is either entered directly or calculated by the device from the entered figure of the bases or base pairs per nucleic acid molecule.
One Cantiague Road, P.
Page 37 BioPhotometer plus — Operating manual 8 Troubleshooting 8. The methods differ mainly in the preprogrammed factor. Mnaual condition of delivery Set-up and initial operation Instruction of user Self test where applicable IQ Certificate. When purifying nucleic acid without a kit, additional contamination risks are posed by the chemicals used in the process, e. Dilution Conversion Calibration report Call up using Enter.
The biophootmeter of the filters are measured against a blank filter “blank A0”. Don’t see a manual you are looking for? Linear and non-linear regression Evaluation procedure of the multi-point calibration. Page 15 BioPhotometer plus — Operating manual 5 Operation 5 5. Nucleic acid purification is a major application in molecular laboratories.
White Paper — Better safe than sorry. The following values represent a pure DNA sample: Using the cursor keys select the desired method.
Eppendorf BioPhotometer® D30 – Instruments, Detection – Eppendorf
BioPhotometer plus — Operating manual 5 Operation 5. Page 1 B Contact us Distributors Technical Service. Have the cuvettes for the measurements available. The inside of the device is a high-voltage area.
Because this is a stray-light measurement the result depends on the geometry of the light path which can differ for the photometers by different manufacturers. Operating manual — BioPhotometer D Microvolume measuring cell for measuring 1.
The light path height of the BioPhotometer plus is 8. Front and rear view Fig. You can modify most parameters and save them as a modified method see Parameter on page In addition to the results the device also displays the absorbance values and some other important details, e.
For specific methods, additional measurements are recorded and displayed as a graph, making impurities easier and quicker to identify. Or point us to the URL where the manual is located. Take advantage of the two-in-one cuvette, decide based on your sample. Further information about the Eppendorf PhysioCare Concept: Your manual failed to upload In touch with life Your local distributor: Users are guided through every step of the assay with short, simple on-screen instructions. Tell us what’s missing.
In addition, all important raw data pertaining to the DNA sample are displayed in table format.
Cell Protein DNA Core Facilities
Eppendorf macro Vis Cuvetteplastic cuvette for measurements in the Vis range, max. To ensure minimum contamination of the sample, it is essential to verify its purity manhal photometric measurements. Press one of the method group keys to return to the method selection and open the next method, if necessary.
BioPhotometer plus — Operating manual 5 Operation 5 5. Damage from improper cleaning of the cuvette shaft. Raw and processed data are clearly presented on the large color screen and can be exported directly from the instrument in a variety of common formats.
The height of the light beam in the cuvette is 1. BioPhotometer plus — Operating manual 4 Installation 4 4. Parameter on page The scattering of the measured values around the calculated calibration line or curve is very large see Section 12, “Calculation”.